Urine Drug Test Kit – MET (Methamphetamine) Rapid Test Dipstick
$6.95 – $20.00
The MET Rapid Test Dipstick is a rapid chromatographic immunoassay for the detection of Methamphetamine in human urine at the cut-off concentration of 300ng/mL.
A discount is applied to this product in your cart when you order 2 or more.
DIRECTIONS FOR USE:
Allow the test, urine specimen, and/or controls to reach room temperature (15-30ºC) prior to testing.
1. Bring the pouch to room temperature before opening it. Remove the Test Dipstick from the sealed pouch and use it within one hour.
2. With arrows pointing toward the urine specimen, immerse the Test Dipstick vertically in the urine specimen for at least 10-15 seconds. Do not pass the maximum line (MAX) on the Test Dipstick when immersing the strip. See the illustration below.
3. Place the Test Dipstick on a non-absorbent flat surface, start the timer and wait for the coloured line(s) to appear. Read results at 5 minutes. Do not interpret the result after 10 minutes
INTERPRETATION OF RESULTS:
NEGATIVE:* Two lines appear. One colour line should be in the control region ©, and another apparent colour line should be in the test region (T). This negative result indicates that the Methamphetamine concentration is below the detectable level of 300ng/mL.
*NOTE: The shade of colour in the test region (T) may vary, but it should be considered negative whenever there is even a faint colour line.
POSITIVE: One colour line appears in the control region ©. No line appears in the test region (T). This positive result indicates that the Methamphetamine concentration is above the detectable level of 300ng/mL.
INVALID: Control line fails to appear. Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for control line failure. Review the procedure and repeat the test with a new Test Dipstick. If the problem persists, discontinue using the Test Dipstick immediately and contact your local distributor.
For further details click here to see the package insert.